Máy đo sóng số TBS1052C
Thông số kỹ thuật máy hiện sóng số Tektronix TBS1052C
|Parameters||Máy hiện sóng TBS1052C||Máy hiện sóng TBS1072C||Máy hiện sóng TBS1102C||Máy hiện sóng TBS1202C|
|Bandwidth||50 MHz||70 MHz||100 MHz||200 MHz|
|Sample Rate||1 GS/s||1 GS/s||1 GS/s||1 GS/s|
|Record Length||20 K points||20 K points||20 K points||20 K points|
đại lý Tektronix | nhà phân phối Tektronix | Tektronix việt nam
Vertical resolution 8 bits
Input sensitivity range
1 mV/div to 10 V/div max. in 1-2-5 sequence with probe attenuation set to 1X
DC gain accuracy
±3.0 % step gain, derated at 0.1 %/°C above 30 °C
Maximum Input Voltage
300 VRMS, Installation Category II; derate above 4 MHz at 20 dB per decade to 200 MHz
1 mV/div to 50 mV/div: ± 1 V
100 mV/div to 500 mV/div: ± 10 V
1 V/div to 5 V/div : ± 100 V
20 MHz (Typ)
1 MΩ ±2 % in parallel with 14 pF ±2 pF
Vertically expand or compress a live or stopped waveform
Acquire sampled values
Captures glitches as narrow as 4 nsec at all sweep speeds.
From 2 to 256 waveforms included in average.
Averages multiple sample of one acquisition interval into one waveform point.
Scrolls waveforms right to left across the screen at sweep speeds slower than or equal to 40 ms/div
Horizontal system – Analog channels
±25 x 10-6 over any >1 ms interval
TBS1202C, TBS1052C, TBS1072C, TBS1102C
2 ns/div to 100 sec/div in a 1-2-4 sequence
Horizontally expand or compress a live or stopped waveform
± 100 nsec
External trigger input
Included on all models
Auto, Normal, Single Sequence
Positive or negative slope on any channel. Coupling includes DC, HF reject, LF reject, and noise reject.
Trigger on width of positive or negative pulses that are >, <, =, or ≠ a specified period of time.
Trigger on a pulse that crosses one threshold but fails to cross a second threshold before crossing the first again.
CH1, CH2, AUX IN, AC Line
DC, Noise Reject, High Frequency Reject, Low Frequency Reject
Trigger signal frequency readout
Provides a frequency readout of the trigger source up to instrument bandwidth.
Time, Amplitude, Screen
32, of which up to six can be displayed on-screen at any one time. Measurements include: Period, Frequency, Rise Time, Fall Time, Positive Duty Cycle, Negative Duty Cycle, Positive Pulse Width, Negative Pulse Width, Burst Width, Phase, Positive Overshoot, Negative Overshoot, Peak to Peak, Amplitude, High, Low, Max, Min, Mean, Cycle Mean, RMS, Cycle RMS, Positive Pulse Count, Negative Pulse Count, Rising Edge Count, Falling Edge Count, Area, Cycle Area, Delay FR, Delay FF, Delay FR, and Delay RR.
Isolate the specific occurrence within an acquisition to take measurements on, using either the screen, between waveform cursors or full record length.
Add, Subtract, and Multiply waveforms
Spectral magnitude. Set FFT Vertical Scale to Linear RMS or dBV RMS, and FFT Window to Rectangular, Hamming, Hanning, or Blackman-Harris.
7-inch TFT Color Display
800 horizontal by 480 vertical displayed pixels (WVGA)
Vectors, Variable persistence, and Infinite persistence.
YT and XY
Input Output Ports
USB 2.0 Host Port
Supports USB mass storage devices
USB 2.0 device port
Rear-panel connector allows for communication/control of oscilloscope through USBTMC or GPIB with a TEK-USB-488
Kensington Style Lock
Rear-panel security slot connects to standard Kensington style lock
100 to 240 VAC RMS ±10%
Power source frequency
45 Hz to 65 Hz (100 to 240 V)
360 Hz to 440 Hz (100 to 132 V)
30 W maximum